This unit is now exactly the same as GS-1100-PRO This page has just been left here to avoid broken links.
|GS-1102-PRO Technical Specifications|
|Input Power (USB)||+ 5V|
|Current Draw||< 200 ma|
|Output Bias Voltage (smooth potentiometer)||0V to 1100V|
|Output Signal (Audio Line Level)||+ 0 - 2500mV|
|Pulse length||100 µs|
|Preamp Gain (adjustable trim pot on bord)||set to ~10%|
|Ripple||< 1 mV|
Includes: USB Cable and male to male 3.5 mm jack plug cable and SMA to BNC Cable
Gamma spectacular outputs an audio signal proportional to the input pulse energy, and is designed to work with sound card spectroscopy software, utilizing the analog to digital capabilities of modern computers.
This unit does not come with a radiation detector. For basic radiation activity measurements, it will work with most GM tubes or pancake detectors, and for Gamma spectroscopy it will work best with NaI scintillation detectors coupled for positive voltage. Please feel free to send us the make and model number of your radiation detector, to confirm suitability.
Any computer with a sound card and mic/line in audio jack port will work. The recommended software is for PC, but it will work perfectly on Mac OS-X with Crossover by CodeWeavers
There are several great programs available for soundcard spectrometry, and most of them are FREE, please see our software download section for the program that suits your need here.